Building a 10 & 24GHz setup to try to beat the 24GHz world record in some weeks withe the “Le Grande Bleue” expedit… https://t.co/sYZ1n1wseR
Performing a cold sky/ambient load measurement to check how performs the new LNA with the @ea3hmj moonbounce setup.… https://t.co/eonHWfJ5u2
I can assure you that the time under the microscope passes at a different speed, it is mesmerizing! #ghz_bands… https://t.co/yOpCoNNw8f
Starting to build the 1st batch of 47GHz LNA’s 🥳🥳🥳
#GHz_bands #hamradio #hamr #rfporn https://t.co/60vgOaxAHS
Today wirebonding practice again, I need warm my wrist again! https://t.co/rysD3EjNZe
Nice evening, tracking the @LucyMission in X-band https://t.co/PmLSv7STXt
The K band down converter used to receive the @NASAWebb the /2 LO is PLL at 11808MHz, and the IF at 2284MHz.… https://t.co/P6ohV2vR7I
Nice carriers from the @NASAWebb in K band at 25900MHz when was xmiting data to @Madrid_DSN at 14mbps.
After finish… https://t.co/bO5rz1JiCs
Doing surgery in my 8510c to add noise figure measurement 🙈
#keysight #RFporn #GHz_bands #hamr https://t.co/kuNrQrhgTO
Yeeeeeees! the divine justice some time works!
Congratulations @Max33Verstappen #FormulaOne #F1Finale
Poor man’s PNA-X project! I want to convert my 8510C in a single-connection multiple-measurements SMMC device. 🙈🙈
Replacing a aluminium bond in a old eprom. The gold wire grips very well over Al. https://t.co/kC9OOL7O5l
The noise figure measurement, sucks. The Y method not is good for very los noise LNA’s, too much uncertainty.
Hola @Madrid_DSN vais a hacer tracking de la #DARTMission ?no la veo en el schedule de la web. Gracias!
Today 47GHz party, calibrating nouse sources and trying to use loads to measure NF. Somebody have experience with t… https://t.co/Kss4G68DnK
Repair technician’s hat again. Today a 10GHz PA for tropo and RS.
#RFporn #GHz_bands #hamradio https://t.co/dN4ZLdf41Y
After 24 years I’ve a SGI Indy at home again! ☺️
It is one of the most pretty computer ever made, I love it! 😍 https://t.co/GEBNS3tv2C
Today repair guy hat. After recap my @Keysight DC power supply, the issue was in a faulty rectifier. Tnx @ea4eoz fo… https://t.co/zMb66YphBv